The water flowing internally through the tip, effectively cools the area and owners assists in removing any debris from the operative site.
This frequency is then amplified and delivered to the scaling tip.
Never clean the handpiece with offline boiling water, chemical solutions, ultrasonic cleaner, or city with wire brushes.
The internal circuitry converts ordinary line voltage to an operating frequency of approximately 18,000.
Thank you for purchasing an Engler Son-Mate II Scaler Polisher engler combination.
Do not lubricate the handpiece while it is on the Micromotor.
(25.4 um to 102 um).A reinforced solid aluminum chassis surrounds the internal circuit board, providing a very durable and reliable unit.Wipe the handpiece clean with an alcohol-soaked soft tissue.East, explorer Europe, Central and South America, Canada, Asia, New Zealand, Australia, and most other countries.Lubrication of the straight handpiece is required at least once a week.Note: If spray time is too short oil may not be propelled into all areas of the handpiece.Lubricate the Handpiece as described in the lubrication section of this manual.Important: We recommend that ultrasonic scalers should not be used by or in close proximity to anyone who has any electronic implant device (pacemaker, defibrillator, or has any neurological problems).Place the Handpiece in an autoclaving pouch and seal it in accordance with the instructions on the pouch.As a result, the tip vibrates offline at this ultrasonic frequency with an amplitude.001.004.Page 17.Web site: please read offline very carefully, if you have any questions or comments, please contact: Engler Engineering Corporation 1099 East 47th Street m, help site: m, page 3.As the bubbles in the lavage are bactericidal, the energy released collapses and destroys the bacterial cell walls.To lubricate, insert the.